I seek advice and opinions from fellow safety minded people.
Our typical product has a metal chassis which is constructed as a Fire
Enclosure (laboratory equipment 61010-1). On the back is a cutout for a cooling
fan/blower mounted in the cutout. The fan is 24Vdc, certified by several safety
agencies, and has a finger guard. This is typical configuration on most
Now, our customers want us to install an externally changeable Air Filter on
the outside surface of our instrument where the fan is mounted to filter dust
particles from the incoming air. We already perform Blocked Vent and Stalled
Fan tests to insure no hazards are caused from the rise in internal
But what type of Fault Testing or Risk assessment needs to be done regarding
the filter? With the filter removed, the instrument passes the construction
requirements for a Fire Enclosure. But with the filter installed and because of
its close proximity to the fan/blower;
1. does this filter have to meet Flammability requirements? Does the
filters have to be certified (expensive)? UL 94 HF-1?, UL 94 HF-2?, UL 900?
How are these ratings/certifications viewed outside of North America? Will
they have to meet local requirements??
2. If the fan/blower is certified and limited energy circuit does the
filter have to have any kind of rating or certification? In other words, with a
certified fan/blower, do I have to consider the fault condition of the fan
failing in a way where it could catch the filter on fire?
3. How is a fault and/or risk assessment performed on an air filter?
4. Are we responsible to consider the hazards from a burning filter when
it is dirty? How would we know what type of contaminates might collect in a
5. If the filter we provide is UL 94 HF-1 and UL 900 rated/certified,
what would stop our customer from replacing it with whatever filter they
wanted? Are we responsible to include a warning label and statements in the
manual regarding this? Example, "Use only Air Filter part number XYZ".
6. Any other suggestions or issues that we are not considering?
Thanks in advance. Have a nice day.
The Other Brian
LECO Corporation Notice: This communication may contain confidential
information intended for the named recipient(s) only. If you received this by
mistake, please destroy it and notify us of the error. Thank you.
This message is from the IEEE Product Safety Engineering Society emc-pstc
discussion list. To post a message to the list, send your e-mail to
All emc-pstc postings are archived and searchable on the web at:
Attachments are not permitted but the IEEE PSES Online Communities site at
http://product-compliance.oc.ieee.org/ can be used for graphics (in well-used
formats), large files, etc.
Instructions: http://www.ieee-pses.org/list.html (including how to unsubscribe)
List rules: http://www.ieee-pses.org/listrules.html
For help, send mail to the list administrators:
Scott Douglas <sdoug...@ieee.org>
Mike Cantwell <mcantw...@ieee.org>
For policy questions, send mail to:
Jim Bacher: <j.bac...@ieee.org>
David Heald: <dhe...@gmail.com>