-------- In message <120112818.2234439.1476263954...@mail.yahoo.com>, Bruce Griffiths wr ites:
>Does anybody know if a 34401A can be adjusted by a competent 3rd party lab, >or are the details of how to adjust the meter not available outside of >Keysight? The service manual has the calibration procedure, and it is all software. -- Poul-Henning Kamp | UNIX since Zilog Zeus 3.20 p...@freebsd.org | TCP/IP since RFC 956 FreeBSD committer | BSD since 4.3-tahoe Never attribute to malice what can adequately be explained by incompetence. _______________________________________________ volt-nuts mailing list -- firstname.lastname@example.org To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/volt-nuts and follow the instructions there.