> the offset of the leontp device from the other clocks has consistently been
> in the 9.5 -10.5 range. since I'm measuring all three sources from the
> same (EL7) computer, I would expect that the offset of the leontp unit to
> converge to be in the close neighborhood of the offsets of the arbiters. It
> has not converged, instead maintaining the ~10ms offset.
The delay column shows the same small delay on all 3 boxes. That pretty much
rules out something strange on the network side.
I would add a few outside clocks. That might help you figure out which clock
is right/wrong. (Both Arbiter's could be setup with the same error.)
You might check the antenna cable delays. (on all 3 boxes) Somebody might
have typed in the wrong number, ms vs ns or km vs feet.
What sort of setup is LeoNTP? Is it triggering on the trailing edge of a 10
ms wide PPS pulse?
These are my opinions. I hate spam.
time-nuts mailing list -- firstname.lastname@example.org
To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
and follow the instructions there.