Thanks Gary, just what I needed, someone with more knowledge of the User's
Guide. Thanks for saving me a bunch of time! I unplugged the console, took
the batteries out and reversed these steps. Boom, back in business. I took
your suggestion of changing the interval. I reset it to 10 minutes.
On Monday, February 12, 2018 at 8:01:12 PM UTC-7, gjr80 wrote:
> Data is great but logs are better :) Quite possible you have corrupt
> station memory (have a look here
> <http://weewx.com/docs/usersguide.htm#html_generated_but_not_updated> in
> the User's Guide), not uncommon when disconnecting/interrupting power to a
> Davis station. If you want to take a punt and don't mind losing any records
> in your station memory, you can clear the console memory using wee_device
> <http://weewx.com/docs/hardware.htm#vantage_clear_console_memory>. If you
> don't want to take such drastic action you can check the logs first, in
> weewx.conf you set debug=1, save weewx.conf then stop/start weeWX. Let it
> run for a couple of archive periods (appears you have a 30 minute archive
> period from your data) then post the log.
> As an aside, if you want an archive interval somewhat less than 30
> minutes, use wee_device to set the archive interval
> <http://weewx.com/docs/hardware.htm#vantage_archive_interval46> as
> required. Note that any use of wee_device must be done with weeWX stopped.
You received this message because you are subscribed to the Google Groups
To unsubscribe from this group and stop receiving emails from it, send an email
For more options, visit https://groups.google.com/d/optout.