On 9/18/2016 5:42 AM, David C. Partridge wrote:
I suspect that you are measuring the analyser phase noise, but it's
rather early in the morning...
I've just redone the measurement without the external attenuator and with 10dB
attenuation set internally to the analyser.
The results are attached.
I attach, if I can get it to work, a plot of three different Tbolts on
my 3048A system.
time-nuts mailing list -- firstname.lastname@example.org
To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
and follow the instructions there.