In December I lucked out buying a HP5065A for a very reasonable price, as is
not tested. Turns out it was not working and I was not able to fix it. Corby to
the rescue. Corby and I have shared many projects over the past, go back over
15 years, as a matter of fact he introduced me to time-nuts. Took Corbe over a
week to find the problem, a defective Rb87 reference cell. Never seen before,
no crack visible but obviously no Rb87. Corby had a replacement and the result
is that it is the third best 5065A he has ever worked with. The logical choice
is the super conversion. Not dramatic improvement because it was already very
Now a step by step work over, including HP mods for later units like replacing
the 74196’s in the synthesizer module with 74LS196. May have been end of life
of the 74196.
Now to the purpose of this post. The A9 module is after 88 a significant change
and Corby sees an improvement against his Maser. We are doing a board and maybe
some other will be interested.
Here are the issues
A is the LT1793 the best choice the time constant is 0.05 seconds with a 10
K resistor and 5 uF Capacitor
B should we add resistors and decoupling on the + - 15 volt op amp supplies
C Gold plating the edge connecter, does any one know a reasonable source,
or is doing it at home an option and if yes, how best way to do so.
time-nuts mailing list -- firstname.lastname@example.org
To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
and follow the instructions there.