If your pages are updating at 10 minute intervals that sounds like the
archive interval being used is 10 minutes (if you are using the SteelSeries
skin that would further support this idea). With a Davis station any
archive interval setting in the console will override the [StdArchive]
archive_interval setting. To confirm what is happening go back and look at
your log from when you last started weeWX, you will probably see something
like this in the start up logging:
Feb 13 13:45:18 stretch22 weewx: engine: The archive interval in the
configuration file (300) does not match the station hardware interval (600).
Feb 13 13:45:18 stretch22 weewx: engine: Using archive interval of 600
If you can't find it stop then start weeWX and have a look at the startup
To fix it, stop weeWX and then use wee_device to set the desired archive
interval in the console
Note, the only effect this will have on gauge-dat.txt is to generate it
every 5 minutes rather than every 10 minutes. Refer to my previous post.
On Tuesday, 13 February 2018 13:51:22 UTC+10, A.J. Burnett wrote:
> Hi Andrew,
> Station type is a VP2. My StdArchive interval is set to 300, which should
> be every 5 minutes, but the gauge-data.txt file only updates every 10
> minutes. My web page only updates at that same 10-minute interval too.
> <scratching head>
You received this message because you are subscribed to the Google Groups
To unsubscribe from this group and stop receiving emails from it, send an email
For more options, visit https://groups.google.com/d/optout.